Untethered, powered orthotics require an actuation system with power supply and control, transmission line and actuator. Fluid power has unmatched force-to-weight and power-to-weight compared to electromechanical systems, but it is unclear if those same advantages hold for small systems in the 10 to 100 W range. A systems analysis approach suggests that a fluid power system will be lighter than an electromechanical system with the same output power and efficiency if the fluid power is run at pressures over about 200 psi. A theoretical analysis of small bore cylinders suggests that eliminating the piston seal will result in a higher efficiency actuator if the clearance gap is small. A demonstration, battery powered electrohydraulic actuator assembled from off-the-shelf components had the force and power suited to a powered ankle orthosis, but is too large and too heavy, suggesting the need to develop custom components.
|Original language||English (US)|
|Number of pages||1|
|Journal||IEEE ... International Conference on Rehabilitation Robotics : [proceedings]|
|State||Published - 2011|
|Event||Rehab Week Zurich 2011 - 2011 IEEE International Conference on Rehabilitation Robotics, ICORR 2011 - Zurich, Switzerland|
Duration: Jun 27 2011 → Jul 1 2011
PubMed: MeSH publication types
- Journal Article
- Research Support, U.S. Gov't, Non-P.H.S.