A high-performance, wide dynamic range, fully-integrated neural interface is one key component for many advanced bidirectional neuromodulation technologies. In this paper, to complement the previously proposed frequency-shaping amplifier (FSA) and high-precision electrical microstimulator, we will present a proof-of-concept design of a neural data acquisition (DAQ) system that includes a 15-bit, low-power Delta-Sigma analog-to-digital converter (ADC) and a real-time spike processor based on one exponential component-polynomial component (EC-PC) algorithm. High-precision data conversion with low power consumption and small chip area is achieved by employing several techniques, such as opamp-sharing, multi-bit successive approximation (SAR) quantizer, two-step summation, and ultra-low distortion data weighted averaging (DWA). The on-chip EC-PC engine enables low latency, automatic detection, and extraction of spiking activities, thus supporting closed-loop control, real-time data compression and/or neural information decoding. The prototype chip was fabricated in a 0.13 \mum CMOS process and verified in both bench-top and In-Vivo experiments. Bench-top measurement results indicate the designed ADC achieves a peak signal-to-noise and distortion ratio (SNDR) of 91.8 dB and a dynamic range of 93.0 dB over a 10 kHz bandwidth, where the total power consumption of the modulator is only 20 \muW at 1.0 V supply, corresponding to a figure-of-merit (FOM) of 31.4fJ/conversion-step. In In-Vivo experiments, the proposed DAQ system has been demonstrated to obtain high-quality neural activities from a rat's motor cortex and also greatly reduce recovery time from system saturation due to electrical microstimulation.
|Original language||English (US)|
|Number of pages||16|
|Journal||IEEE transactions on biomedical circuits and systems|
|State||Published - Jun 1 2020|
Bibliographical noteFunding Information:
Manuscript received November 6, 2019; accepted January 25, 2020. Date of publication February 6, 2020; date of current version May 27, 2020. The technology portion of this work was supported in part by National Science Foundation CAREER Award #1845709, in part by the National Institutes of Health under Grant R01-MH111413-01, in part by the MnDRIVE Program at the University of Minnesota, and in part by a gift fund from Nerves Inc., Dallas, TX, USA. The experiment portion of this work was supported in part by Defense Advanced Research Projects Agency under Grants 1198 HR0011-17-2-0060 and N66001-15-C-4016, and NIH R21-NS111214-01. This article was recommended by Associate Editor Dr. Danilo Demarchi. (Corresponding authors: Jian Xu; Zhi Yang.) Jian Xu, Anh Tuan Nguyen, Tong Wu, Diu Khue Luu, and Zhi Yang are with the Department of Biomedical Engineering, University of Minnesota, Minneapolis, MN 55455 USA (e-mail: firstname.lastname@example.org; email@example.com; firstname.lastname@example.org; email@example.com; firstname.lastname@example.org).
- Data acquisition
- EC-PC spike processor
- delta-sigma ADC
- dynamic range
- electrical microstimulation
- high precision
PubMed: MeSH publication types
- Journal Article
- Research Support, Non-U.S. Gov't
- Research Support, U.S. Gov't, Non-P.H.S.
- Research Support, N.I.H., Extramural